|Standard Warranty (2-year)||NA:BBX24||
|Extended Warranty (4-year)||NA:BBX24+EW-BBX||
The original Bullet Blender enables you to homogenize, disrupt, or lyse up to 24 small (300 mg or less) tissue or cell culture samples at a time. Load the samples into standard polypropylene microcentrifuge tubes (1.5 to 2.0 ml), then place them in the Bullet Blender. The "bullets" in the instrument vigorously strike all of the tubes and homogenize your sample in minutes.
The Bullet Blender prepares up to 24 samples in just minutes. All of the samples are automatically homogenized in the same manner to minimize sample-to-sample variability.
There are no probes or blades to cause cross contamination or sample loss, which also means there are no parts to clean between runs. The sample tubes are kept closed and self-contained during homogenization.
The instrument uses very little power due to the unique, patented design, so there is very little heat build up in the instrument or inside the sample tubes compared to other brands of homogenizers. It uses a DC motor to agitate the individual tubes. The Bullet Blender is designed for reliability and durability, with few moving parts and a very low rate of motor wear.
Simply place your sample, beads and buffer in standard tubes, and load the tubes into the Bullet Blender. Tune the extent of disruption by adjusting the run time and speed. Afterwards, place the sample tubes in a centrifuge to separate the lysate. No special holders are needed. There are no clamp plates nor locking rings to fiddle with.
Bullet Blender units come with a 30 day money back guarantee and a 2 year warranty.
from Plant Samples
from Animal Samples
from Microbial Samples
Genomic DNA samples in agarose gel stained with ethidium bromide.
Lanes 1-4 = Tobacco leaf DNA
λ = phage λ Hind II digest
Lanes 5-8 = Rice leaf DNA
Total RNA gel from mouse tail snips.
RNA integrity number = 8.5
Data courtesy of Madhu Prasad @ U. Michigan
E. coli sample.
SDS Page gel stained with coomassie blue.
wc = whole cell. sf = soluble fraction.
Left two lanes control. Right two lanes induced.
|Homogenizer Type||Bead Mill|
|Sample Throughput||24 samples|
|Sample Volume (min)||30µl|
|Sample Volume (max)||900µl|
|Usage / Placement||Benchtop|
|Noise Level||90 dB|
What is the RPM range of the Bullet Blender?
Next Advance does not disclose the RPM specifications for the Bullet Blender. The reason they do this is because the mechanism of action for the Bullet Blenders (striking) is very different from the mechanism of action for other laboratory bead mill homogenizers, which almost exclusively act via shaking. Direct comparisons with different bead mills from different manufacturers based on RPMs, including using protocols from other instruments based on RPM, are not possible.
The best we can tell you is that the speed setting "1" is a speed which would be good for a moderately vigorous mixing, and the maximum speed is the most power you will get from the instrument. The specific RPMs which correspond to the numbers on the dial are not documented by Next Advance.
If you have questions pertaining to the processing power of the Bullet Blender, please contact us for more information.Next Advance 2013-06-09